CellDrop Brightfield Beads and CellDrop Fluorescence Beads at ~1×106 beads/mL are available for brightfield and fluorescence applications respectively. The beads should be stored at 2 – 8oC in an airtight container. The Fluorescent beads should be protected from light.
The Brightfield app counts cells that have bright centers surrounded by dark rings. The Brightfield beads, when properly focused, appear as thick dark rings with a small bright center and crisp edges. These beads work well in the Brightfield app, and they will be counted as live cells in the Trypan Blue app. The recommended protocols to use for brightfield beads are below:
Trypan Blue App
DeNovix Fluorescent Bead fluoresce in both green and red channels. The fluorescent beads are a 50/50 mixture of red-fluorescing beads and green and red fluorescing beads with a dimmer red fluorescence. These beads may be counted in any app with fluorescence channels. The AO/PI app counts green and red fluorescing samples and is most appropriate to use with the DeNovix Fluorescent Bead Standard.
The recommended protocol for fluorescent beads is below:
AO / PI App
- Vortex bead suspensions prior to use.
- Ensure that the arm is in the down position, and launch the appropriate app.
- Clean the sample surfaces if there is visible debris in the preview image.
- Select or create a protocol.
- With the arm down, dispense the sample aliquot into the measurement chamber using the groove on the lower sample surface as a pipetting guide.
- Note: The volume of the sample depends on the protocol settings for the chamber height. The required volume is displayed on the Count button.
- Adjust the focus and exposure.
- When the beads have settled and are no longer moving, tap the Count button.